Manometers MP KIMO MP KIMO MP Data Sheet ( KB) For detailed information please refer to: [email protected] · Environnement SA. This is the MP Thermo-Anemo-Manometer. It is an easy to use meter which can take complex measurements. The MP is designed to measure Air. KIMO Instruments portable thermo-anemo-manometer with backlit digital display. Functions: pressure, temperature, air velocity and airflow with Pitot tube, CO.
|Genre:||Health and Food|
|Published (Last):||14 July 2004|
|PDF File Size:||11.25 Mb|
|ePub File Size:||5.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
Air velocity with Pitot tube: Wire probes with Smart-plus system Wire probes with Smart-plus system are automatically recognized when connected to the instrument. The ”Probe” menu only appears when probes or module are connected.
KIMO MP P USER MANUAL Pdf Download.
This menu allows to view probe information plugged to C2, Module, C1 or wireless connections. Probe display Available information are: On enabled mode, the probe is connected, the measurement is carried out and the value is displayed.
On disabled mode, the probe is connected, the measurement is not carried out and the value is not displayed. With Pressure function, you can access to ,p200 sub-functions – Hold – see Air velocity – Config. This sub-function allows to compensate for any long-term drifts of the sensing element by a manual adjustment of the zero. Once pressing Autoz key, the zero is readjusted.
This function can also be automatically performed by using the solenoid valve function. For others measurement modules, self-calibration is performed by disconnecting the two pressure inlets of the sensor, then by pressing Autoz key.
If K factor is selected, you must enter value. Probe Hold Alarms Params Veloci.
You can access this ikmo selecting COmax with the access key CO max The CO is measured on an adjustable period, the maximum value measured in this period is called CO max.
When CO peak is selected, the period is diplayed 30 seconds by default. When the countdown is finished, the CO max is displayed. To modify the period, press Period with the access key. If the probe is connected for more than 1 minute, measurement screen appears and indicates the concentration and places it on a bargraph. If the probe has been just connected, the screen indicates the remaining time of warm up then the measurement screen appears, indicates the concentration and places it on a bargraph.
Several actions are possible during the measurement: When the measurement is fixed, press Min Max to access jp200 maximum and minimum values of the measurement. From this screen, press Measure to back to measurement.
Kimo MP 200 H Thermo Anemometer Manometer Air Velocity Pressure Sensor
Use the mimo to modifiy the threshold value then press Validate to validate alarm threshold and kmio to measurement screen. Alarm threshold corresponds to the right value on the bargraph. The beep repetition frequency will increase accordong to the increase of detected gas concentration, and the display will blink at the same frequency.
Configuration – Params Parameters – Avg. Average – Rec Recording Veloci. Press 1x in order to select HOLD function: Press 2x in order to select Min-Max function: Numbers of selected points and parameter for which calculation is carried out, are displayed For adding a new measuring point to this calculation, press OK to confirm.
If you click on average icon, max. If you iimo to see all values, select Visu. Select Start with access key for launching measurement.
MP – KIMO – PDF Catalogs | Technical Documentation | Brochure
You must enter duration: The numbers of points is displayed. Press Ok for launching limo. You can view each measuring points if you click on Visu. Configuration If you use thermocouple probes, you must enter type into the Configuration sub-function.
KIMO User manual MP 200 Manometer
Configuration sub-function allows to: Click on Esc to return to previous screen. Select compensation It is possible kijo modify the value of the compensation in temperature.
It is thus necessary to enter the real temperature of use to obtain more precise results. If Other is selected, you must enter a value. When the solenoid valve is enabled, it runs every minute. You can choose between a planned or a continuous dataset.
Jimo capacity of the instrument is up to 8, points or 50 datasets. Create or launch a continuous dataset Select dataset A continuous dataset can be carried out using MP and is composed klmo several dated measuring points. The operator can choose an automatic or a manual dataset, with an instant value or an average. This datasets can’t be set using Datalogger Software. For measurement launching, click on OK with the access key.
The number of points selected and the parameter are displayed. To save your dataset click on Save with the access key. Enter dataset time and interval of time between mp20 measurements by selecting Period with access key. Select Start for dataset launching. Launch a planned dataset A planned dataset is composed of several locations. For each kiml, the operator can enter a theorical value and a tolerance for the parameter to be controlled.
Planification must be made via the software. Preview of tables of points of datasets Dataset summary You can display tables of points of datasets performed on the device.
Go to Recording menu. Click on OK to validate. Summary screen of selected dataset is displayed. From this screen, you can: Click on Mesure to display values table of selected dataset. From this screen you can: Repeat this operation for the month, year, hour and kumo. Battery When battery indicator flashes it is recommended to change the batteries: Remove the front part at the back of the instrument.
In regards of Quality insurance norms, we recommend that the instruments are checked once a year. It’s a community-based project which helps to repair anything.